Designing the inside of the crown
This step covers the design of the inside of the crown/inlay – the part that will be in contact with the preparation. This part of the construction, and the parameters involved, are crucial for proper fitting. In case you are not happy with the fitting of the produced part, it's here where you should look for possible parameter changes.
The cement gap
The thickness of the cement gap can be set using the slider . The default value corresponds to the setting in the Ceramill Mind record card. Changes made here only apply for the current design (in contrast to changes in the Ceramill Mind record card, which then apply as new standard values with the dentist selected). Move the slider  to alter the thickness of the cement gap – changes are adapted in real time, the effects of the action can therefore be seen immediately.
A click on the text “Add another area”  activates an additional slider, which can be used for marking a second cement gap area and defining its thickness.
Areas of the crown base can be individually defined using the brush function. To define an area click on the “Brush” button  and mark the required area on the crown base with a left click.
The slider “From the margin”  defines the start of the cement gap in relation to the preparation margin in millimetres. If, for example a value of 1 mm is selected, the surface  in an area of 1 mm around the preparation margin will not have a cement gap. In the same way the end of the cement gap can be specified “from the top”  using the slider to define a surface  without a cement gap in the occlusal area of the crown. The arrow  shows the path of insertion again. If this arrow is highlighted red, the die has undercuts in the area of the preparation margin using the selected path of insertion.
By unfolding the „Additional spacing“ expander, you will get access to options  and  which allows you to add additional spacing.
- This can be defined separately for axiale and radial direction by using the slider .
- Additional spacing is also applied in the area where there is no cement gap
- Additional spacing is applied in addition to the cement gap in area where there is a cement gap
Desiging the crown border
The second tab of the „Crown Bottoms“ dialog gives you access to parameters that define the shape of the crown border. The graphic  gives an illustration of meaning of the sliders [1-4].
 Defines the horizontal crown border width. Typically, material properties enforce certain limits here; e.g. for Zirconia a common minimum value is 0.2mm.
 Defines the length of the angled part of the border. This may commonly be set to zero.
 Defines the angle of the angled part.
 Defines an additional vertical border. Commonly this is also set to zero.
Dealing with undercuts
The third tab in the „Crown bottoms“ dialog gives you access to undercut-related options and milling parameters.
Undercuts in the preparation are blocked out by default, unless you specifically choose not to do so, by checking checkbox .
A certain (small) angle may be applied for blocking out undercuts, as defined by . By default, a value of zero is used, meaning that undercuts are blocked out „straight down“. Slider  defines an „untouchable zone“ around the preparation margin, where the crown is never blocked out, even if it should be (due to undercuts being present above). We will further elucidate the effect of this option with an example. Consider the two images below (for this tooth, we have intentionally set an incorrect insertion axis to illustrate the problem).
The graphic on the left shows the behavior with „Don't block out zone near prepline“ set to 1mm. Observe how the upper part is blocked out, but as soon as the untouchable zone is reached, the crown bottom will go back to the preparation (note that it is not possible to produce this kind of restoration with Ceramill Motion). The graphic on the right shows behavior with „Don't block out zone near prepline“ set to zero. As you can see, the crown will not be in contact with the preparation around the margin. In both cases, the result will be non-optimal and the restoration will need manual work (either by manual milling, or by adding ceramic) for a proper fit. It is up to the operator to decide which option may be better in the particular clinical case.
Any parameters you change in this screen will not be applied in real time, as these are computing-intensive operations. Click „Apply“  to see the effect of your changes. To visualize undercuts which remain (e.g. due to usage of „Don't block out zone near prepline“), click the „Show undercuts“ button .
When done, click „Next“ to proceed to the next step.